Thank you for choosing our steam sterilizers.
Prior to operating this instrument, please read the operations
manual carefully and follow
all installation instructions.
Caution: Be sure connect the exhaust tube and discharge container before
start the sterilizer.
Henan Forever Medical Co., Ltd.
Wanda Center,Jinshui District,
Subject to technical changes
|Electrical supply||220-240 VAC||110-100 VAC|
|Chamber (mm, diameter/depth)||Ø230×360 mm|
|Power (VA)||1100 VAC|
|Sterilization temperature (°C)||121 °C||134 °C|
|Overall dimensions (mm)||605 (width)x445(height)x400(depth)|
|Net weight (kg)||30 kg|
|Sterilization time||121 °C||30 min.|
|134 °C||10 min.|
- Position the device on a plane surface with minimum capacity 60 Kg.
- The sterilizer should be placed on a level worktable.
- Improper water level in the chamber could cause a sterilizer
- Leave at least 10 cm between the device rear part and the wall. The
clearance required to open the door is 40cm.
- Position the autoclave at such a height as to make it possible for
the operator to check the whole sterilization chamber and carry out
the normal cleaning operations.
- The room where the device is installed must have sufficient
- Do not install the autoclave near washing basins, taps, etc. where
it is likely to be splashed.
- Do not install the sterilizer nearby a heat source.
- Do not lean on the door when it is opened.
- Do not place trays, papers, fluid containers, etc. on the
- Check the label on back panel of sterilizer to verify voltage
rating for the unit. Failure to connect
the autoclave to an appropriate power supply could result in damage
to the unit, and electrical shock to personnel.
- Plug power cord into a properly polarized and grounded receptacle
rated. A dedicated circuit only used for the sterilizer is
Open the door and remove all the inner contents for unpacking.
Connect the power cord to an outlet of the appropriate voltage.
Connect the draining tube to the vent and the end of the draining
tube is fixed into the discharge container (preferably a metal
container; if it is a plastic container, make sure that the
container is filled with a part of cold water to prevent high
To power on, press the main switch, the machine turns on the LED.
Note: If the door is open, the display will show "do"; When the
door is closed, the display will show "Ld".
Preparation of sterilization materials
For the most effective sterilization and to preserve the sample,
please follow below:
- Clean instruments immediately after use.
- Treat the instruments by ultrasonic cleaner.
- Residual chemicals left over after cleaning and disinfecting
process may damage and corrode parts of the autoclave, always rinse
off the instruments using distilled water.
- Follow instrument manufacturer's guidelines and recommendations for
handling and cleaning instruments prior to sterilization.
- Check the manufacturer' s instructions as to the proper procedure
for sterilizing of each item.
- Arrange the samples of different materials on different trays or
with at least 3cm of space
- Clean and dry instruments thoroughly before placing them into tray.
Close the exhaust valve by turning it clockwise to the maximum
Fill with the chamber with distilled water (or deionized water).
Select the program temperature.
After selecting program, the materials to be sterilized can now be
placed on the tray placed inside the chamber using the tray handle.
After the instruments are loaded, you may close the chamber door.
The display will show the current internal temperature.
Caution: You must turn the door handle to the maximum position.
Start the sterilization program.
After the Start button is pressed, the sterilizers will begin the
The draining tube must be always connected to the vent and the end
of the draining tube must be fixed into the discharge container.
End of cycle
After the sterilization is complete, you will hear a beep and the
screen will show "End".
Then you can open the door to remove the sterilized items.
After the screen shows End, if the gauge is pointer different than
0 bar, do not open the door until it down to zero-scale position.
After each cycle check the water level of the chamber.
|Program||Type||Max. Load (kg)||Max. Load|
per tray (kg)
|121 °C / 134 °C||Unwrapped solid material||2.4||1.2|
|E2||Chamber overheating||Check temperature sensor connection.|
|E4||Heating overtime||Somewhere is leaking. Check the door seal.|
|E6||During operation, the door has been opened or the door micro switch
is damaged.||Confirm that the door handle has been closed to the maximum
Check or replace the door micro switch
|Monthly||Clean the door seal.|
|Weekly||Clean the chamber, the trays and|
|Every Year||Replace the door seal.|
Clean Chamber, Trays, Door Seal, chamber filter and Tray Rack.
Remove the trays and tray rack from the chamber. Clean trays, rack
and inside of chamber with mild soap.
Rinse the trays, rack and inside of chamber with a smooth cloth and
Examine gaskets for possible damage.
Clean gasket and mating surfaces with a damp cloth.
Note: Do not use bleaching agents or any abrasive materials /
substances in chamber. Failure to comply may result in damage to
the chamber and/or other components.
Caution: To prevent burns, allow unit to cool before cleaning
gaskets and mating surfaces.
Under normal circumstances the chamber door does not require
adjustments. However, if the seal
fails (resulting in steam leaking from the front of the chamber),
you may use the spanner tool to
tighten the door seal.
Open the door and Insert the spanner tool in the gap beneath the
plastic cover; use the spanner to grip the adjusting nut.
Turn the nut counter clockwise. This will tighten the sealing
Turn the nut until the sealing plate is tight. If the door knob is
too tight, you may also turn the nut clockwise to loosen it.
Replacement of the door seal
Open the chamber door. Remove the door
seal ring carefully by hand.
Clean the door seal ring carefully with a
smooth cloth with distilled water.
Moisten the new seal with medical
disinfectant or isopropyl alcohol.
Insert the new seal and press in sequence
Caution: Please ensure the chamber and the
door have cooled prior to replacing the seal ring.